After Installing a home weather station, there are numerous pieces of equipment in the weather station that require regular maintenance and calibration checks. The user must never overlook this section, as the weather station’s accuracy is entirely dependent on weather station maintenance.
1. Terminal maintenance of home weather station
The terminal must be properly maintained. It should not be exposed to the sun. Sensors attached to the terminal must be maintained away from heat and cold sources. The terminal’s batteries of the weather station should be replaced as needed, especially if the brightness becomes dim. If the power goes out, it should still be possible to copy files to and from the drive.
2. Sensor System Maintenance of Home Weather Station
How can you keep your weather station sensors working properly and avoid expensive repairs when they break down? Follow these simple weather station maintenance tips to keep your weather station running smoothly and save yourself a lot of grief.
A visual or aural inspection at low wind speed must be carried out every 1 month. If the anemometer emits an unusual buzzing or rotation, there could be issues with the anemometer bearings. Check for cracks or damage to the anemometer wires and make sure the anemometer is correctly routed and free of tension.
External Temperature / Humidity Sensors
To clarify, we suggest that you clean your rain gauge several times a year. Dust and debris may accumulate in the collector bucket, which affects the operation of the tipping mechanism. For cleaning, Remove the cap from the base of the rain catcher to wash it. Using water and a light liquid detergent, gently wipe the tip bowl and funnel. Wash the dish well.
Solar Radiation Sensors
Dirt on the diffuser at the top of the solar radiation sensor of the weather station can degrade its accuracy. Dust with a gentle dry brush as frequently as possible. If necessary, wipe around the rim with a wet cloth or cotton swab.
The solar radiation sensor should be cleaned at least once a month. The level of the sensor may be verified using the leveling bubble found on the radiation sensor’s base plate, and the screws must be adjusted if required. Compressed air or a soft bristle brush should be used to remove any dust or debris from the sensor head.
3. Power source Maintenance Of Home Weather Station:
If the weather station is fueled by solar energy, the panel must be examined every three months. The tilt and direction of the orientation must be examined and corrected. The glass of the solar panel should be examined more frequently and cleaned with a soft, slightly damp cloth since dirty glass can lower the solar panel’s power output.
If the battery of the personal weather station is not adequately maintained, two issues arise: first, corrosion, and second, voltage. To avoid these issues, the battery needs to be cleaned regularly.
To avoid corrosion, the terminals should be lubricated on a regular basis. If the corrosion is severe, the battery’s terminal connections must be changed as soon as possible. If a problem with the battery’s voltage is detected, the voltage must be tested with a voltmeter between the battery’s poles.
Why did my personal weather station stop working?
Cause: If the terminals of the weather station are not receiving enough power, this might happen. It’s possible that the power cord is loose. Check the power supply as well as the cable connection. Make that the power supply is firmly hooked into the outlet. If the battery of the personal weather station is not adequately maintained, two issues arise: first, corrosion, and second, voltage. To avoid these issues, the battery needs to be cleaned regularly.